The device is compatible with standard "low profile" 0.1 mL PCR tubes. 0.2 mL tubes are not compatible with our device. Additionally, standard PCR optical tape alone may not adequately prevent sample evaporation. We suggest adding at least 25 uL of mineral oil on top of your sample before sealing with optical tape.
However, we highly recommend using Biomeme’s Go-Strips, as they are specially designed for use with our device. If you run into issues using tubes other than Go-Strips, please contact email@example.com for troubleshooting tips.